

Articles about bioresonance therapy


Distinctive peculiarities of the devices «BIOMEDIS M» and «BIOMEDIS»

The distinctive feature of the devices of this series in comparison with analogues is following: the method of direct digital synthesis (DDS) of frequency is used for generation of treating signals; it allows getting high accuracy of frequency setting.
The functional scheme of the synthesizer DDS is showed on the picture 1: its main components include a stacker for value of a phase (phase accumulator), a tool for converting phase value into amplitude (it is usually a memory device, containing table with function sine values) and the DigitaltoAnalogue Converter (DAC).
The scheme DDS generates sinusoidal signal, having given frequency. The output frequency is determined by two parameters: the clock signal and the binary number written in frequency register. The binary number written in the frequency register is available at the input of the phase accumulator. If the memory device, containing tabular function sine values, is used, the phase accumulator defines the address (which corresponds to instant phase value) and sends it to the input of the memory device; thus we get the current value of amplitude in digital form at the output of the memory device. Then the DAC converts the digital value into the corresponding value of voltage or current. For generating sine wave with fixed frequency a constant value (increment of phase which is determined by the binary number written in frequency register) is added to the value saved in phase accumulator with every impulse of clock signal. If value of increment is great, the phase accumulator will quickly run the table of sinuses contained in the memory device, and signal of frequency will be high. If the value of increment is low, the phase accumulator will need more steps in order to run the whole table of the memory device, and that's why the signal of frequency will be low at the output.
The DigitaltoAnalogue Converter (DAC) made on the same crystal with the generation's scheme of digital counts (DDS) represents the complete integral DDS synthesizer.
Instant phase value of continuing sinusoidal signal changes cyclically within the diapason, ranging from 0 to 2?. The phase's value generates in digital form. The transfer's function which is contained in the counter allows realizing the continuing cyclic change of phase's value in the synthesizer DDS. Imagine sinusoidal oscillations in the form of vectors rotating around the wheel (picture 2) in order to understand how generation takes place! Each point on the wheel corresponds to certain point of sine wave. While vector rotates around the wheel, the value of the sine of angle is an output signal. One turnover of the vector, having constant speed, provides generation of a sine wave's period.
The phase accumulator generates angle's values, having equal increments; the value contained in the phase accumulator corresponds to certain point of the circle.
The phase accumulator represents a moduloM counter; the value of the counter increases each time it gets a clock impulse. The value of increment is specified by the binary number M. This number determines the value of phase's increment with each clock impulse; this number actually determines the quantity of counts skipped while rotating around the wheel. The greater the step is, the more quickly the phase accumulator overflows and the shorter the period of sine wave is. Total number of possible phase values is determined by phase accumulator's resolution; and then the total number of possible phase's values determines frequency resolution of the synthesizer DDS (n). The accumulator will overflow after 228 cycles (clock impulses) for 28bit phase accumulator, if the value M = 0000…0001. The accumulator will overflow after 2 cycles (the minimum of cycles which meet required criteria by Nyquist), if the value M = 0111…1111. This relationship is described by the following simple formula:
fout = М x fc/2^n
where:
fout is frequency of output signal of DDS;
М is the binary number which determines frequency of signal;
fc is frequency of clock signal;
n is length of the phase accumulator.
If the value M is changed, the frequency is immediately changed at the output of the synthesizer; the signal doesn't have any interruption in this case. Loop settling time doesn't take place here, as it is incurred in the case of phaselocked loop. The number of counts per once cycle decreases in case of increasing output frequency. Since the theory of counts requires at least two counts per cycle for complete reconstruction of output signal, the maximum frequency of the DDS synthesized signal is fc/2. But the practice shows that the frequency of synthesized signal is actually limited by lower value; this condition provides improvement of the synthesized signal's quality and easing of its filtering. When the signal of constant frequency is generated, the code at the output of the phase accumulator increases linearly; it corresponds to linear analog ramp signal.
The memory device, containing tabular values of sinus, is used for converting of phase accumulator's output code into instant amplitude values. Insignificant bits of 28bit code are eliminated; we get a 10bit code at the output of tabular memory device; the code is sent to DAC. Since the sine wave has symmetrical nature, the DDS synthesizer saves only the data of onequartercycle of the sine wave. The tabular memory devices generates the complete cycle of the sine wave by reading data forward and then back.
DDS AD9832 with 32bit phase accumulator is used in the devices "BIOMEDIS M" and "BIOMEDIS; it allows setting frequency with the following accuracy: 8 MHz / 2^32= 0,00186264514923095703125 Hz within the diapason of frequencies ranging from 0 to 4000000 Hz.
Now let examine the principle of generating treating signals by similar devices that don't use DDS. The elaborators have utilized a simple way: the way of dividing the signal of clock cycle into integer number, while using inserted timers of processor or through organizing of program divider.
For example, we want to generate a signal, having certain frequency Foutput or cycle Toutput through microprocessor. We can divide the clock frequency Fclock only into certain integer number n and get Foutput /n at the output.
Let see what we get in this case. If the frequency of clock generator is 5 MH:
N 
Foutput 
643 
7776,05 
644 
7763,975 
645 
7751,938 
646 
7739,938 
647 
7727,975 
648 
7716,049 
649 
7704,16 
650 
7692,308 
651 
7680,492 
652 
7668,712 
653 
7656,968 
654 
7645,26 
655 
7633,588 
656 
7621,951 
657 
7610,35 
658 
7598,784 
659 
7587,253 
660 
7575,758 
661 
7564,297 
662 
7552,87 
663 
7541,478 
664 
7530,12 
665 
7518,797 
666 
7507,508 
667 
7496,252 
668 
7485,03 
669 
7473,842 
670 
7462,687 
671 
7451,565 
672 
7440,476 
673 
7429,421 
674 
7418,398 
675 
7407,407 
676 
7396,45 
677 
7385,524 
678 
7374,631 
679 
7363,77 
680 
7352,941 
681 
7342,144 
682 
7331,378 
683 
7320,644 
684 
7309,942 
685 
7299,27 
686 
7288,63 
For example, we need the frequency = 7334 Hz (it is the second frequency by Rife in case of Chemtrail detox). As it is seen in the table above, the frequency is not included in the table. The nearest frequencies are following: 7342,144 and 7352,941 Hz. Or for example, we need to get the frequencies 7848 and 7847 Hz (the first frequencies of Ecoli). The frequencies are greater; it means that the step is larger. So, we fail again.
The step of the greed approaches 0,1 Hz in case of output frequencies which are equal to about 720 Hz.
N 
Foutput 
4143 
1206,855 
4144 
1206,564 
4145 
1206,273 
4146 
1205,982 
The step of the greed approaches 0,1 Hz in case of output frequencies which are equal to about 720 Hz.
N 
Foutput 
6921 
722,439 
6922 
722,3346 
6923 
722,2302 
6924 
722,1259 
6925 
722,0217 
6926 
721,9174 
6927 
721,8132 
6928 
721,709 
6929 
721,6048 
6930 
721,5007 
6931 
721,3966 
6932 
721,2926 
6933 
721,1885 
6934 
721,0845 
6935 
720,9805 
6936 
720,8766 
6937 
720,7727 
6938 
720,6688 
6939 
720,5649 
6940 
720,4611 
6941 
720,3573 
6942 
720,2535 
6943 
720,1498 
6944 
720,0461 
6945 
719,9424 
6946 
719,8388 
6947 
719,7351 
It means that the lower the frequency of generating signal is, the narrower the step of the greed is. The accuracy of 0.01 Hz is provided only in the diapason ranging till 100 Hz.
The next distinctive peculiarity of the devices "BIOMEDIS M" and "BIOMEDIS" is usage of voltage increase converter. Why is it important?
The supply voltage of the whole device should be within of certain limits for stable work of the microprocessor and graphic indicator. The voltage needed for processor should usually range from 2.7 to 5.5 volts; the voltage range for indicator should be even narrower. Thus, if two AA batteries are used for supply, the voltage range for each of them is lower than 2.7 volts. As a result, the device fails to work (or it works improperly: the processor works but the indicator doesn't reflect data; intensity of radiation falls because it is in quadratic dependence on supply voltage). The absence of the voltage increase converter doesn't allow replacing the batteries AA by rechargeable accumulators that provide saving money in case of prolonged usage of the device. The devices "BIOMEDIS M" and "BIOMEDIS" contain voltage increase converters, having the efficiency coefficient of 98% that allows supporting the voltage on the supply line of the device at the level of 5V in case of input diapason ranging from 0,8 to 6 V. The working time of the device increases till replacing of supply elements in this case; moreover, the user can be sure that stability of radiation's intensity is provided. 



